After upgrading the Equalizer software to v99, all measurements stopped working and the LED on the Equalizer is blinking red, indicating a loss of measurement from the NL P1 port. I suspect a (critical) bug in the v99 version of the software. Tested with Sagemcom T210-D ESMR5 (3-phase) meter.
I'm having the same issue. I reported this issue last week to Easee's Support and they're working on a fix. I was told v99 was only intended to be pushed to the HAN (Norwegian) version of the Equalizer, but instead it was pushed to _all_ Equalizer versions, including the P1 variant. In the BeNeLux Facebook group someone has received a firmware update v103 that fixed the issue for him, but I do not see this version mentioned yet on the changelog page nor in the installer app or cloud environment.
v103 has been pushed to my Equalizer this morning which seems to resolve the case for me.
I'll contact support and ask them to push v103 to mine as well. Thanks
I have exactly the same issue and setup. Equalizer is reporting a very high current 80000A on phases and is blinking red.